Yes, you can also take blood samples from the palm, forearm, upper arm, thigh or calf. *Always check with your doctor before performing alternative site tests.
Food, medication, illness, stress and exercise can affect blood glucose levels. Capillary blood in the fingertips reflects these changes faster than capillary blood from other sites. Thus, when testing blood glucose levels during or immediately after a meal, physical exercise or any other situation, take a blood sample from your finger only.
．Pre-meal or fasting state (more than 2 hours after the last meal).
．Two hours or more after taking insulin.
．Two hours or more after exercise.
The meter stores the 450 most recent blood glucose test results along with respective dates and times in its memory. The meter memory is activated when the meter switched off.
Push "Main button".
will appear on the display. Push Main button again, and the first reading you see is the last blood glucose result along with the date, time and the measurement mode.
Push "Main button" to recall the test results stored in the meter each time you press. After the last test result, push “Main button” again and the meter will turn off.
1. Push and hold "Main button"
|2.||Push "Main button" to review 14-, 21-, 28-, 60- and 90- day average results stored in each measuring mode in the order of Gen, AC, and then PC.|
Exit the meter memory.
|NOTE:||．Any time you wish to exit the memory, keep pressing Main button for 5 seconds or leave it alone for 3 minutes. The meter will switch off automatically.
．Control solution results are NOT included in the day average.
．If using the meter for the first time, “---“displays when the test results are recalled or the average result reviewed. This indicates that there is no test result in the memory.
If you want to turn off an alarm, find the alarm number by pushing Set in the setting mode, push “Main button” to change from “ON’ to “OFF”.
When the alarm goes off, the meter beeps and automatically turns on. You can push “Main button” to silence the alarm and insert a test strip to begin testing. If you do not push “Main button”, the meter will beep for 2 minutes and then switch off. If you do not wish to do the test at this time, push “ Main button” to switch off the meter.
There are seven (7) speaking volume options to choose from. Push "Main button" until the desired speaking volume appears. To confirm your selection, push Set.
Volume 0 indicates that the speaking function is turned off, and “ ” will not be displayed during testing. Volume 1 to 7 indicates speaking volume from low to high, and “ ” will be displayed throughout the testing.
You can use the meter with an USB cable and the Health Care Software to view test results on your personal computer. To learn more about the Health Care Software System or to obtain an USB cable separately, please contact your local customer service or the place of purchase for assistance.
Connect the cable to a USB port on your computer. With the meter switched off, connect the other end of the USB cable to the meter data port. "USb" will appear on the meter display, indicating that the meter is in communication mode.
To transmit data, follow the instructions provided with the software. Results will be transmitted with the date and time. Remove the cable and the meter will automatically switch off.
Replacement lancets and test strips are distributed through major distributors to your nearest stores and pharmacies, Please contact your local customer service or the place of purchase for assistance.